Enteric Viral Screening
EV002 – EasyScreen™ Enteric Viral Detection Kit
EV002 (100RXNS) & EV002-HT (500RXNS)
For the detection of (i) Norovirus GI (ii) Norovirus GII (iii) Astrovirus (iv) Rotavirus A (v) Sapovirus (vi) Enterovirus (vii) Bocavirus (viii) Adenovirus and (ix) Adenovirus (40 + 41 only) in faecal specimens. The multiplexed PCR is performed over three PCR wells and includes an Extraction Control (EC) and Extraction Process Control (EPC) within panels, to monitor RNA integrity and PCR inhibition, in the following layout:
|Detection Channel||Reaction Mix A||Reaction Mix B||Reaction Mix C|
|Channel 1||Norovirus GII||Sapovirus||Adenovirus Hexon Universal*|
|Channel 2||Extraction Process Control|
|Extraction Process Control|
|Channel 3||Astrovirus||Norovirus GI||Bocavirus|
|Channel 4||Rotavirus A||Enterovirus||Adenovirus 40/41*|
*Assay detects both Adenovirus universal sequence and Adenovirus 40/41 in order to distinguish ingested respiratory causing Adenovirus.
The EasyScreen™ Viral Detection Kit is designed to be used in conjunction with EasyScreen™ Sample Processing Kits and include all reagents required for amplification of 100 reactions or 500 reactions in HT format.
For more information about the EasyScreen™ Workflow and Compatibility with Nucleic Acid Extraction and Real-Time PCR platforms visit our EasyScreen™ Workflow page.
Typical Results Obtained from Primary Patient Samples
Results of a clinical study comparing results from EasyScreen™ Enteric Viral Detection Kit to EIA from faecal samples.
|EIA positive||EasyScreen™ Positive|
*The previous version of this kit did not contain an Enterovirus target
More clinical data can be found on our Independent Clinical Results Page.
Examples of Typical Results
Clinical specimens (stool samples) were assayed with the EasyScreen™ Enteric Viral Detection Kit and typical results for each channel (Target Mix A) are represented below.